Portable Anesthesia Machine
Screen: 8.4 '' LCD Display Screen
Suitable: Adult & Child
Mode: pneumatically powered and electrically controlled system
Working Mode: Closed; Semi-Closed; Semi-Open
Circuit: Breathing circuit integrated standards
Flowmeters: 5 Tubes Flowmeters: O2:0.1~10L/Min, N2O:0.1~10L/Min; Air: 0.1~10L / Min
Trolley: Fitted with 4 nos anti-static rubber castors; two of which are lockable for braking and eaay maneuverability with foot operated brake provisions
Drawer unit: 2 drawers totally extractable
Gas requirement: Medical oxygen and nitrous oxide with a pressure ranging from O2: 0.32~0.6MPa ;
NO2: 0.32 MPa to 0.6 MPa.
safety valve: <12.5 kPa
Respiratory Rate: 1~99bpm
oxygen concentration in the mixed gas N2O/O2: > 21%
Oxygen Flush: 25~75 L/min
modes of ventilation: A/C, IPPV, SIPPV, IMV, SIMV, PCV, VCV, PEEP, SIGH, PEEP, MANUAL
PEEP: 0 ~ 20 hPa
Ptr: -10 ~ 20 hPa
IMV frequency: 1 ~ 20 bpm
I/E Ratio: 8:1 ~ 1:10, Has the inverse ratio ventilation
Tidal Volume: 0 ~ 1500 ML
Inspiratory Plateau: 0~1s
O2 Concentration: 21%~100%
Sigh: one deep breath per 70 ~ 120 controlled respirations, the inspiration time is 1.5 times of the setting point
Maximum safety pressure: ≤ 12.5 kPa
Pressure limit range: 0 ~ 6.0 kPa
Airway pressure alarm: Audible and visual and with yellow and red color indicating
Upper: 0.3~6.0 kPa ±0.2 kPa
Tidal volume alarm: Audible and visual and with yellow and red color indicating upper
alarm: 50 to 2000ml, lower alarm: 0~1800ml
Oxygen concentration alarm: Audible and visual and with yellow and red color indicating
upper alarm: 21% ~100%; lower alarm:10%~80%
Power Supply Alarm Ac/dc power supply are after failing to send out alarm immediately
Alarm Time: keep >120s
The airway pressure continues to be higher than 15 hPa ±1 hPa for 15s±1s, then the machine will raise an audible alarm, the pressure will be displayed in red and the continuous high pressure red alarm wording is displayed on the screen of the anesthetic respirator.
Ambient temperature: 10 ~ 40oC
Relative humidity: no higher than 80 %
Atmospheric pressure: 860 hPa ~ 1060 hPa
Power requirement: ~220-240V, 50/60Hz, 40VA, to be well grounded.
Air source requirement:medical oxygen and laughing gas with a rated pressure ranging from 0.3 to 0.5MPa.
Attention: the AC power supply used for the anesthesia machine must be well grounded.
Sweep up and pay attention to official account.
Nanchang beiaote Medical Technology Co., Ltd. Gan ICP B 15006598 - 1 this website supports the visit of IPv4 / IPv6! Technical support: 300.cn Nanchang